Figure 6. Location
brackets, access door, and compartments.
a suitable type hoist and utilizing
hook-up. The hook-up for one 12-volt
the four lifting eyes (fig. 12) which
and two 6-volt batteries is the same
a r e permanently attached to the
base of the test stand. Remove the
6-volt batteries A and B are replaced
wooden skid-type base, a above, and
with one 12-volt battery.
position the test stand in place on
the floor of the selected location.
Caution: When the battery leads
Note. Normally the test stand need not
are not connected, insulate the loose
be secured to the floor for depot installa-
ends of the leads by taping or by
tion, however, if need arises for this re-
other suitable means or disconnect
the leads and remove from the bat-
of mounting holes for securing the test
tery compartment to prevent acci-
stand in position.
dental grounding of the test stand
Caution: The test stand is
shipped without oil in the gear case
of the varidrive assembly. Do not
(4) Depot Installation.
operate the test stand until the
(a) Test stand (figs.
1, 2, and 4). Select
gear case has been filled to the
a location for the test stand which
proper level with lubricating oil
is properly ventilated, dry, and not
subject to extreme heat, dripping
moisture, or exposed to dirt. The
location should be where hazardous
1. Venting of the air intake and ex-
processes are not being performed.
haust are not required for depot
Lift the test stand off the floor using
installation; however, be certain