(b) Short machine's terminals with
(g) If meter does not respond prop-
a piece of wire or a straightened paper clip,
erly in (d) and (e) above, clean battery contacts,
observe indicator lamp, and squeeze handle
tighten battery connections, and repeat (d) and
sharply. Lamp should light.
(h) If meter still does not respond
(c) Remove short, observe indicator
lamp, and squeeze handle. Lamp should not
properly, replace battery in accordance woth (b)
and (c) above, and repeat (d) and (e) above.
(d) If lamp responds properly in (b)
(i) If meter still does not respond
and (c) above, test set is ready for use.
properly, organizational maintenance personnel
should dispose of galvanometers in accordance
(e) If lamp does not respond prop-
with local material accounting procedures.
erly in (b) and (c) above, organizational mainte-
nance personnel should dispose of test set in
(2) Test set M51.
accordance with local materiel accounting proce-
(a) Unpack test set and wipe clean
with soft, dry cloth.